|Stock:||Yes||Shelf Life:||3 Years, 3 Years|
|Instrument Classification:||Class I||Safety Standard:||/|
|Product Description:||Polyester Swab/ Rayon Swab/ Dacron Swab||Color:||White|
|Application:||TOC Cleaning Validation,reovery Study|
Polyester TOC Swab,
Disposable TOC Swab,
Cotton Disinfection Wipe Stick
Can be customized medical disposable polyester-cotton swab medical consumables disinfection wipe stick
HCY(Huachenyang)TOC Cleaning Validation Swab(kit) was designed to simplify sampling as part of a cleaning validation protocol. The swabs and vials are further cleaned using a proprietary process to make these kits excellent for use in total organic carbon(TOC)analysis. Lot Coded for traceability and quality control.
①. Supporting independent combined packaging:
Organized, convenient, and safe transportation method;
From the storage area to the sampling point to the swabs and vials in the laboratory, the possibility of contamination is very small.
Total organic carbon content <50 µg/L, <50 ppb, to ensure that the swab itself does not interfere with the test data;
Double-layer and double-layer woven polyester head, which can carry pollutants; knitted structure in the sampling process. This is to maximize sample collection and release into the diluent to provide an excellent recovery rate;
Separate handle with a notch, which can put the head into the vial for the smallest processing and sampling;
100% polypropylene handle to ensure no other contaminants and has excellent chemical resistance.
③, Sample bottle
Total organic carbon content (<10µg/L, <10 ppb), to ensure that the swab itself does not interfere with the test data;
The pre-washed borosilicate glass bottle has an integral cap and diaphragm design to facilitate sampling and analysis.
Diaphragm thickness -1.34 mm (0.053")
Polyethylene cladding protects the diaphragm surface from contamination before analysis.
The oblique shoulder design reduces the area of the cotton swab head and reduces the total organic carbon content.
Approved by TOC instrument manufacturer for use.
The cleanroom is suitable for the identification and traceability of vial samples.
Double-layer bag, packed in polypropylene box compatible with cleanroom.
1. Please refer to your written SOP sampling instructions.
2. Dip the cotton swab into the solution.
3. Sampling area.
4. Remove the cap of the sample vial.
5. Place the cotton swab to jam the tip of the cotton swab into the vial, and the notch touches the top of the vial. Avoid touching the swab head.
6. Close the lid.
7. Mark the sample.
|Product name:||Polyester swab|
|Breakpoint||Can be broken or can not be broken|
|Shelf Life||3 years|
|Intended Use||TOC Validation/Specimen Collection|
|Application||TOC cleaning Validation,reovery study|
|Certificate||CE, ISO Approved|
|Instrument classification:||Class I|
Contact Person: edward